Foxboro d/p Cell Transmitters have been the standard of process (a) See Optional Capsule Materials (13A Series Only) for span limits of 13A4 (Monel capsule. The Foxboro® brand Models 15A, 13A, and 13HA pneumatic d/p Cell transmitters measure differential pressure and. Download Manual Model. Regarding This Manual. • This manual should be provided to the end user. • The contents of this manual are subject to change without prior.
|Published (Last):||8 October 2011|
|PDF File Size:||13.3 Mb|
|ePub File Size:||6.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
Foxboro 13A, 13HA, 15A Pneumatic D/P Cell Transmitter
Share to Twitter Share to Facebook. Most transmitter calibration done is a five point calibration. In most calibration done, you will be doing either a bench shop calibration — A bench calibration is a procedure where the device is calibrated at a foxbori bench using calibration devices to simulate the process, — or a field calibration where the actual process is used.
That is, adjusting one has an effect on the other.
Foxboro 13A,13HA,13DMP Pneumatic Differential PressureTransmitters
How to Calibrate Smart Transmitters. CalibrationDP Transmitters. Readings are taken for both increasing and decreasing input values and the corresponding transmitter output values are recorded Step 5: The graph below illustrates the correlation between input and output values.
For your application, this could be modified slightly. The potentiometer has a maximum of 20 turns between minimum and maximum resistance therefore turning the ZERO or Msnual screws clockwise or counter-clockwise for 20 turns will cause the potentiometer to be at either maximum or minimum.
You are now ready to calibrate the DP transmitter. So once your equipment is well setup, power it up and pressurize the high port of your DP transmitter.
Models 15A, 13A, and 13HA Pneumatic d/p Cell® Transmitters
Whether you are doing a bench calibration or a field calibration, the low port of the DP transmitter cell is vented to the atmosphere as shown in the connection diagram above and the high port of the DP transmitter connected to a pressure source e. All 113a reading obtained will be the as found readings.
Connect all the equipment needed for the calibration exercise in the appropriate manner. Note that these screws each connect to a variable resistance potentiometer and can be turned indefinitely.
That is the potentiometer is of a type that once fully adjusted clockwise or counter clockwise the screw may continue to turn without further varying the resistance for either direction.
Record the current reading in m A which will be your first data point. To calibrate an instrument involves checking that the output of the foxoro instrument corresponds to given inputs at several points throughout the calibration range of the instrument. For example if the pressure source is a hand pump, you can easily control the pressure applied to the DP cell.
Then connect them according to your connection diagram. However, if you are doing field calibration that requires the use of the actual process pressure, you will need a pressure regulator in conjunction with a pneumatic calibrator to help you control the pressure applied to the DP cell.
Foxboro 13A, 13HA, 15A Pneumatic D/P Cell Transmitter | Cascade Automation
A typical DP cell transmitter calibration diagram is shown below:. Newer Post Older Post Home. If you calibrate the DP transmitter before first testing and recording the as found data, the history of manua device performance data will be lost.
Record and put down the following can easily be sourced from transmitter nameplate: